Measures glucose value of extreme hematocrit range, like neonate, elder, anemia patient.
Before-and-after meal markers allow you to understand how food affects your blood glucose.
Capillary, venous, arterial and neonatal whole blood are applicable.
Eliminates the needs for manual coding and delivers better accuracy than no coding.
With this Validus Technology™, Rightest™ strips always deliver high accuracy and precision in readings.
The innovative side-insert test strip allows users to avoid contact with the blood sample
How it Works
1. Insert a GS700 test strip in the socket of the meter. The calibration will be done automatically.
2. Place your blood sample against the edge of sample entry of the strip.
3. After 5 seconds, the test result will appear.
4. Rotate and pull up the test strip from the meter.
|Coding||Auto coding calibration|
|Enzyme / Measurement technology||GDH / Electrochemical sensor|
|Blood sample source||Capillary, venous, arterial and neonatal whole blood|
|Minimum sample volume||0.75 microliters (0.00075c.c.)|
Glucose concentrations ≦ 200mg/dL, HCT range 20 – 70%
Glucose concentrations > 200mg/dL, HCT range 20 – 60%
|Measuring range||10 – 600mg/dL (0.6 – 33.3 mmol/L)|
|Meter dimensions (L x W x H)||82.0 x 45.0 x 15.5 mm (3.22 x 1.77 x 0.61 inch)|
|Display area (L x W)||34.0 x 27.5 mm (1.09 x 1.34 inch)|
|Weight (with battery)||43.0 ± 5 g|
|Power supply||1 CR2032 battery|
|Memory capacity||500 blood glucose test results with date and time|
|Operating temperature||6 – 44℃ (43 – 111°F)|
|Operating relative humidity||10 – 90%|
|Meter storage conditions||-10 – 60°C (14 – 140°F)|
|Test strip storage conditions||4 – 30°C (39 – 86°F), < 90% relative humidity|
95% of the measured glucose values shall fall within either
Glucose concentrations < 5.5mmol/L (100mg/dL), difference within ± 0.83mmol/L (15mg/dL)
Glucose concentrations ≥ 5.5mmol/L (100mg/dL), bias within ± 15%
Glucose concentrations < 5.5mmol/L (100mg/dL), SD < 0.28mmol/L (5mg/dL)
Glucose concentrations ≥ 5.5mmol/L (100mg/dL), CV < 5%